EFSEN UV & EB TECHNOLOGY is the European Centre for calibration of EIT instruments and is certified to calibrate according to EIT standards. EFSEN has a calibration facility at our location in Holte, Denmark and we have been trained by EIT to calibrate and issue certificates according to EIT standards. We normally calibrate instruments every Wednesday and return them the following day, provided that no larger repair is needed. Instruments arrived at EFSEN on Tuesday at the latest, will be calibrated the same week.
Please contact our calibration department at email@example.com, firstname.lastname@example.org or email@example.com for any queries and to make sure that calibration department isn’t paused, due to our masters are in the US for calibration.
Please note that EIT and all EIT certified labs will stop service and calibration of the PowerPuck II (PP2-2000) with serial connection by end of December 2022. Last day for calibration of PowerPuck II serial instruments is December 21nd, 2022.
Service and calibration of the PowerMap I and UV Map Plus has been extended to end of June 2023, due to EIT facing delivery problems of components for building PowerMap II.
For further information please contact firstname.lastname@example.org
Please note: Calibration department is closed in week 3-5 all weeks included. Last calibration day before 3 weeks closure is January 11th, 2023. Instruments should arrive no later than January 10th, 2023, to be calibrated before closure.
We calibrate the following EIT instruments
Please Note: Calibration department is closed in week 42 and 43. Last calibration day before closure is October 12th, 2022. Instruments should arrive no later than October 11th, 2022 to be calibrated before closure.
Calibration at EFSEN UV & EB TECHNOLOGY
Calibrations are always performed on an arc lamp, unless another UV source is agreed. If your primary UV source is a microwave lamp, we can offer to calibrate on a Heraeus Fusion lamp to ensure more accurate nominal values.
We calibrate standard high-power instruments (UVA, UVB & UVV range: 100mW/cm2-10W/cm2. UVC range: 10mW/cm2-1W/cm2). We do not calibrate Lower Power (LP) instruments and midrange instruments. UVA, UVB and UVC filters are calibrated on a Hg source and UVV filters are calibrated on a Ga source as standard. All filters can be calibrated on an Hg source upon request.
Call for Calibration
EFSEN call all instruments for calibration before they are due for their biannual calibration. Reminders are sent out by e-mail to the responsible person. If you would like your instrument added to this free service, please send an e-mail to Julie Boesen or Lisbeth Vestergaard Grove with the following information: